Question? Just ask us!
Free Encyclopedia of Building & Environmental Inspection, Testing, Diagnosis, Repair
ENVIRONMENTAL HAZARDS - INSPECT, TEST, REMEDY
AIR CLEANER PURIFIER TYPES
AIR FILTERS for HVAC SYSTEMS
AIR POLLUTANTS, COMMON INDOOR
AIR QUALITY IMPROVEMENT STRATEGIES
ALLERGEN TESTS for BUILDINGS
ANIMAL ALLERGENS / PET DANDER
ANIMAL or URINE ODOR SOURCE DETECTION S
ATTORNEYS and EXPERT WITNESSES
BIBLIOGAPHY ENVIRONMENTAL HEALTH, MOLD, IAQ
CAR MOLD CONTAMINATION
CARPET DUST IDENTIFICATION
CARPET MOLD CONTAMINATION
CAT DANDER in BUILDINGS
COMBUSTION GASES & PARTICLE HAZARDS
CONDENSATION or SWEATING PIPES, TANKS
CPSC Indoor Air Pollution Book Online Copy
DUCT SYSTEM & DUCT DEFECTS
EMERGENCY RESPONSE, IAQ, GAS, MOLD
FLOOD DAMAGE ASSESSMENT, SAFETY & CLEANUP
FLOODS IN BUILDINGS-mold
GAS EXPOSURE EFFECTS, TOXIC
HUMIDITY CONTROL & TARGETS INDOORS
HOUSE DUST ANALYSIS
HOUSE DUST COMPONENTS
INDOOR AIR QUALITY IMPROVEMENT GUIDE
MILDEW REMOVAL & PREVENTION
MOISTURE CONTROL in BUILDINGS
MOLD GROWTH on SURFACES, TABLE OF
MYCOPHOBIA, STAINS MISTAKEN for MOLD
NOISE / SOUND DIAGNOSIS & CURE
ODORS GASES SMELLS, DIAGNOSIS & CURES
OZONE for MOLD OR ODORS
RADON HAZARD TESTS & MITIGATION
SAFETY HAZARDS GUIDE
SICK HOUSE IAQ QUESTIONNAIRE
SMELL PATCH TEST to Track Down Odors
STAIN DIAGNOSIS on BUILDING INTERIORS
VENTILATION in BUILDINGS
Digital camera settings for microphotography: This article explains the optimum settings to use when taking digital camera photographs through the microscope. The purpose of this paper is to help microscopists photograph, store, and use digital microphotographs using modern digital cameras and transmitted-light microscopes. Discussion focuses on selection of cameras adapt well to microscope eyepieces or trinocular heads, and on camera settings and procedures to obtain best quality photographs. We also discuss image resolution and size settings and make recommendations. Cameras used as examples in this paper include the Nikon Coolpix series 990, 995, and 4500.
Green links show where you are. © Copyright 2014 InspectApedia.com, All Rights Reserved.
[The photograph at page very top is one of our earliest attempts at photographing Aspergillus sp. using lacto phenol cotton blue stain. Even a beginner can obtain very good microphotographs with just a little care. The photograph shown here is of one of our stereoscopic microscopes in use for making digital photographs.
The techniques discussed in this paper work well with digital cameras and any type of microscope, low power stereo zoom to high power forensic polarized light microscopes.]
PAAA 2005 Symposium, University of Tulsa, Tulsa Oklahoma - June 2-5, 2005 - updated 02/23/2009
The Nikon Coolpix 900-series and the 4500 camera can take wonderful photos on fully automatic, but to make photomicrographs you may have more success if you make use of some of the manual camera settings as follows (for the Coolpix 990. Find and make the comparable settings in the 850, 950 and 4500 menus). Most other digital cameras will have a similar set of camera setting options but they may be organized around a different menu system. Find and adjust the equivalent settings on your particular camera, using the guidelines listed below.
In general the "automatic" settings of the camera work pretty well - it has general modes of "Automatic", Manual, Program, and Shutter - set by holding Mode (Func 1) button on camera top and turning the thumb wheel - your camera should start in "Automatic".
The camera has a number of image size (3:2, XGA, VGA, FINE) settings and within XGA and VGA you also specify the resolution (Basic, Normal, Fine, and "HI" the last of which makes a huge TIFF file you don't normally want). Image size refers to the intended final display or print size. Resolution refers to the level of data compression in the stored image file. Larger size and higher resolution mean larger image files (and more disk space consumption).
We usually shoot at XGA-Fine which will let us print a high-resolution 8x10 photo if we want a printout, but keeps the disk file sizes down to a manageable 200-300K bytes. You could shoot at XGA-Fine with good result but if you ever want to crop and zoom in on a particle in your photo you may not like the quality. If you want to save disk space and if you never plan to zoom in or enlarge your images, shoot at VGA-Fine which will produce excellent but smaller photos. Remember however that often you may want to enlarge a portion of a photomicrograph to see detail or for publication. You can't do much enlarging of a "VGA" image.
To email photos to someone you'll want to use a photo editing program to make a smaller, lower-resolution image to get the file size down to about 30K. (Folks who have the CD-ROM from the Laurentides Spore Camp received down-sized images which you'll see don't stand up to close scrutiny nor to any zooming.)
Some experts argue that one need not bother to keep images of higher resolution than the resolution capacity of the printer to be used for final images. Because we use images for reference and identification (email-exchange) purposes, often with images cropped and enlarged, and because we rarely print large images, the choice of image size is not reduced to printer resolution.
In sum: for image library purposes and printing up to 8x10 you do not need to store high resolution, large image size files - which wastes disk space. However if you need to crop and enlarge a portion of a photographic image you will want to shoot at higher resolutions and perhaps sizes. If in the microscope we know in advance I'm going to want to crop and enlarge we shoot at fine or high resolution image sizes. we have not required use of uncompressed TIFF type images for these purposes.
Turn the camera on to "Manual" mode" Not to confuse with the "Automatic" overall MODE setting above, to use the Coolpix through the microscope eyepiece we shoot with the ON-OFF DIAL in "Manual" mode (as opposed to fully Automatic) - set the dial that turns the camera on to "M" rather than "A".
Press the "Menu" button on the back of the camera (above the display screen) and then using the round thumb button to scroll and select, and the "picture taking" button to select each setting (there are shortcuts which I'm leaving out for simplicity), choose the following settings:
The *** items are helpful for better photomicrographs but are not essential.
On the back of the camera the bottom left button "M focus" press repeatedly until you see the little Tulip Flower in the upper right display. Now on the upper right of the camera back adjust the zoom (Ranges between "wide angle" and "telephoto") until you see the tulip flower turn yellow - which means you are in "Macro Lens" mode. (On the 850 just turn the dial to "CSM" mode.)
Macro mode and auto focus can often focus the camera to produce a better image than a human can do manually. However if the object of interest lacks strong lines and contrast, the camera may have trouble focusing. Alternatively, you may wish to be sure the focus concentrates on specific features or ornamentation. For these purposes, set the camera to infinity mode, and then observing the image in the camera LCD, you can fine-tune the camera focus by adjusting the microscope stage height slightly up or down as needed.
On the back of the camera bottom center button, press until you see a circle with a
Green link shows where you are in this article series.
Frequently Asked Questions (FAQs)
No FAQs have been posted for this page. Try the search box below or CONTACT US by email if you cannot find the answer you need at InspectApedia.
Questions & answers or comments about digital camera settings when taking photos through a microscope
Check the FAQs just above, try the search box just below, or if you prefer, post a question or comment in the Comments box below and we will respond promptly.
Search the InspectApedia website
HTML Comment Box is loading comments...
Technical Reviewers & References
Related Topics , found near the top of this page suggest articles closely related to this one.