Question? Just ask us!
Free Encyclopedia of Building & Environmental Inspection, Testing, Diagnosis, Repair
InspectAPedia ® Home
ENVIRONMENTAL HAZARDS - INSPECT, TEST, REMEDY
AIR CLEANER PURIFIER TYPES
AIR POLLUTANTS, COMMON INDOOR
AIR QUALITY IMPROVEMENT STRATEGIES
AIRBORNE PARTICLE ANALYSIS METHODS
ALLERGEN TESTS for BUILDINGS
ASBESTOS IDENTIFICATION IN BUILDINGS
BACKDRAFTING HEATING EQUIPMENT
BLACK MOLD, TOXIC & ALLERGENIC
BLEACHING MOLD, Advice about
BOOKSTORE - ENVIRONMENTAL
CADMIUM in the HOME
CARBON MONOXIDE - CO
CARPETING & INDOOR AIR QUALITY
CAT DANDER in BUILDINGS
CELL PHONE RADIATION
CHEMICAL CONTAMINANTS in WATER
COMBUSTION PRODUCTS & IAQ
DIRECTORY of MOLD / ENVIRONMENTAL EXPERTS
DUST SAMPLING PROCEDURE
EMERGENCY RESPONSE, IAQ, GAS, MOLD
EMF ELECTROMAGNETIC FIELDSRE
ENDOCRINE DISRUPTERS at BUILDINGS
FLOOD DAMAGE ASSESSMENT, SAFETY & CLEANUP
FLOOR TILE ASBESTOS IDENTIFICATION
FUNGICIDAL SPRAY & SEALANT USE
GAS EXPOSURE EFFECTS, TOXIC
HEATING OIL EXPOSURE HAZARDS, LIMITS
HOUSE DUST ANALYSIS
HOUSE DUST COMPONENTS
HUMIDITY CONTROL & TARGETS INDOORS
INDOOR AIR QUALITY IMPROVEMENT GUIDE
LAB PROCEDURES MICROSCOPE TECHNIQUES
LEAD POISONING HAZARDS GUIDE
LEGIONELLA LEGIONNAIRES' DISEASE
LIGHT, GUIDE to FORENSIC USE
METHANE GAS SOURCES
MILDEW in BUILDINGS ?
MOISTURE CONTROL in BUILDINGS
MOLD ACTION GUIDE - WHAT TO DO ABOUT MOLD
MOLD CONSULTANTS / INSPECTORS
MOLD DETECTION & INSPECTION GUIDE
MOLD EXPERT, WHEN TO HIRE
MOLD RELATED ILLNESS GUIDE
MSDS Material Safety Data Sheets
MVOCs & MOLDY MUSTY ODORS
NOISE / SOUND DIAGNOSIS & CURE
ODORS GASES SMELLS, DIAGNOSIS & CURE
OIL, HEATING, EXPOSURE HAZARDS, LIMITS
OIL HEAT ODORS & NOISES
OIL SPILL CLEANUP / PREVENTION
PET ALLERGENS / PET DANDER
PET STAINS & MARKS in BUILDINGS
PLASTIC ODORS-SCREENS, SIDING
PLUMBING SYSTEM ODORS
PVC - VINYL BUILDING PRODUCTS
RADON HAZARD TESTS & MITIGATION
SAFETY HAZARDS GUIDE
SAFETY HAZARDS & INSPECTIONS
METHANE GAS HAZARDS
SEPTIC SYSTEM ODORS
SEWAGE BACKUP TEST & CLEANUP
SEWER GAS ODORS
SMELL PATCH TEST to Track Down Odors
STAIN DIAGNOSIS on BUILDING EXTERIORS
STAIN DIAGNOSIS on BUILDING INTERIORS
SULPHUR & SEWER GAS SMELL SOURCES
UFFI UREA FORMALDEHYDE FOAM INSULATION
URETHANE FOAM Deterioration, Outgassing
VINYL CHLORIDE HEALTH INFO
VOCs VOLATILE ORGANIC COMPOUNDS
WATER ODORS, CAUSE CURE
This article explains how to use EMF or ELF measuring instruments when performing electromagnetic field (EMF) or electro-magnetic radiation EMR measurements either by engaging a professional or by consumers using low-cost instruments which measure EMF exposure levels in gauss or milligauss.
We discusses sources of error and variation in EMF measurements and we review and make suggestions for using several low-cost EMF measurement devices to determine the instantaneous electromagnetic field exposure.
Green links show where you are. © Copyright 2014 InspectApedia.com, All Rights Reserved.
For position-insensitive equipment, a single reading is usually provided, directly in mG.
For position sensitive equipment you'll find an enormous range of response depending on the angle and direction in which you hold the measurement device.
While this type of instrument is more work to use (see calculations below and in spread sheet) it provides more clear indication of when you're approaching a field. Some instruments do not provide a reading directly in mG and you'll have to simply record the "raw" measurements and to convert them later.
Some readers may be confused between two common terms used to describe the strength of an electromagnetic field: Gauss and Milligauss. Both of these terms measure the same effect, but at different ranges of strength level. It's simply a matter of moving the decimal point, as we demonstrate in our photographs above.
In our first photo (above left) the ELF EMF meter has been set to the Gauss range and we are measuring a field strength of 0.03 Gauss with our meter touching a small power transformer.
In our second photo (above right) the same ELF EMF meter has been set to its more-sensitive Milligauss range and we are measuring 25 milligauss.
Because usually the electromagnetic field strength around residential properties is very weak, most measurements will probably be made in the milligauss range. 0.03 Gauss is the same as 30 Milligauss.
Our instrument, more accurate in the lower milligauss range, shows that we are actually measuring 25 milligauss, a number which was rounded up to 0.03 Gauss when the scale was shifted.
Definitions of Gauss and Milligauss for Measuring Electromagnetic Fields - EMF
Gauss is a measurement scale used to measure the strength of an electromagnetic field (EMF). The gauss level measured in a given location depends on the strength of the source of the EMF and the distance from the source. The measured field strength of an EMF falls off quite rapidly with distance (field strength declines as the square of the distance from the source).
When electrical current flows through a wire (such as in an electrical distribution wire or inside of an electric motor) an electromagnetic field is produced.
Power lines, local electrical wires, air conditioning motors, computers, TVs, hair dryers, toasters, all devices that use or transport electricity will produce an electromagnetic field.
Usually the strength of these fields is low (Electric fan 1 mG, TV 1mG, chain saw 2.5 mg(?) at chest height, electric stove 6 mG, older electric blankets before circuit redesign 20 mG, and under a distribution line for electrical power transmission, 2-20 mg depending on the line height, KVA rating, and actual level of usage at the time of measurement.
If there is a concern for measuring exposure or possible exposure to electromagnetic fields, it's critical that we have a correct understanding of the levels of exposure that are being examined.
For example, one of our readers informs us that that manufacturer of his pacemaker recommends avoiding exposure to electromagnetic fields stronger than 5 gauss. (EMF generated from small electric motors (such as a chain saw) measured with the saw held in normal operating position may be as much as 2.6 Gauss at the user's chest.)
When recording EMF measurements be certain that you've got the right order of magnitude.
1 Milligauss (mG) = 10-3 Gauss, or 0.001 Gauss
Published sources provide lots of examples of producing and measuring electromagnetic fields at a known gauss level. For example, using a 1.5V battery and a 150 Ohm resistor to make a long straight wire circuit, the EMF field strength when measured at 0.1m (about 4") from the wire will be about 0.2 milligauss (mG).
To compute the actual point measurement, each of these numbers, once converted to mG, must be squared, the three squares added, and the square root taken of the sum. This is because the measurement scale is not linear, so a direct raw average would be incorrect.
In the EXCEL worksheet which we provide
Following good procedure and using instruments properly are two steps towards making accurate, repeatable EMF measurements. But because the signal transmission for RF sources such as radio, TV, or cell towers, the load on a power transmission line is not under control of an individual property owner, and because the EMF strength varies as the power transmission line load varies, it is important to have an idea of that condition as well when attempting to characterize EMF exposure at a specific location.
In contrast, EMF measurements are quite accurate and repeatable at other EMF sources such as close to electrical appliances and service entry cables.
The information provided here is for research and study purposes. The author makes no representation of unique expertise on this topic, other than having field experience in EMF measurement, having studied technical literature and having conversed with other experts and authors in the field for a number of years.
Continue reading at EMF CANCER SCARE or select a topic from the More Reading links shown below.
Suggested citation for this web page
Green link shows where you are in this article series.
Frequently Asked Questions (FAQs)
No FAQs have been posted for this page. Try the search box below or CONTACT US by email if you cannot find the answer you need at InspectApedia.
Use the "Click to Show or Hide FAQs" link just above to see recently-posted questions, comments, replies, try the search box just below, or if you prefer, post a question or comment in the Comments box below and we will respond promptly.
Search the InspectApedia website
HTML Comment Box is loading comments...
Technical Reviewers & References