Question? Just ask us!
Free Encyclopedia of Building & Environmental Inspection, Testing, Diagnosis, Repair
InspectAPedia ® Home
ENVIRONMENTAL HAZARDS - INSPECT, TEST, REMEDY
ACOUSTICAL SEALANT CHOICES
AGE of a BUILDING - how to determine
AIR BYPASS LEAKS
AIR LEAK DETECTION TOOLS
AIR LEAK MINIMIZATION
AIR LEAK SEALING PROCEDURE
AIR POLLUTANTS, COMMON INDOOR
ALLERGENS in BUILDINGS, RECOGNIZING
ANIMAL ALLERGENS / PET DANDER
APPLIANCE EFFICIENCY RATINGS
ARCHITECTURE & BUILDING COMPONENT ID
ASBESTOS CEILING TILES, Asbestos-Containing
ASBESTOS CEMENT ROOFING
ASBESTOS CEMENT SIDING
ASBESTOS DUCTS, HVAC
ASBESTOS-FREE INSULATION MATERIALS
ASBESTOS IDENTIFICATION IN buildings
ASBESTOS FLOORING HAZARD REDUCTION
ASBESTOS FLOOR TILE IDENTIFICATION
ASBESTOS FLOORING REMOVAL GUIDE
ASBESTOS LIST of PRODUCTS
ASBESTOS MATERIAL REGULATIONS
ASBESTOS MATERIAL REGULATIONS Update
ASBESTOS MATERIAL REGULATIONS, OSHA
ASBESTOS PHOTO GUIDE to Materials
ASBESTOS REMOVAL, Amateur, Incomplete
ASBESTOS REMOVAL CERTIFICATIONS
ASBESTOS REMOVAL, Wetting Guidelines
ASBESTOS RISK ASSESSMENT
ASBESTOS TESTING LAB LIST
ASBESTOS UNDER the MICROSCOPE
ROOF ICE DAM LEAKS
BASEMENT HEAT LOSS
BUCKLED FOUNDATIONS due to INSULATION?
BUILDING NOISE DIAGNOSIS & CURE
CATHEDRAL CEILING INSULATION
CEILING FINISHES INTERIOR
CEILINGS, DROP or SUSPENDED PANEL
CEILINGS & WALLS, PLASTER TYPES
CERAMIC TILE FLOOR, WALL
CERAMIC TILE, ASBESTOS in?
CHIMNEY INSPECTION DIAGNOSIS & REPAIR
CHINESE DRYWALL HAZARDS
DEW POINT CALCULATION for WALLS
EFFLORESCENCE SALTS & WHITE DEPOSITS
FIBERGLASS INSULATION MOLD
FLOOR TILE HISTORY & INGREDIENTS
FLOOR TILES ASBESTOS
FLOOR TYPES & DEFECTS
HEAT LOSS in BUILDINGS
HOUSE DOCTOR, how-to be
HUMIDITY LEVEL TARGET
ROOF ICE DAM LEAKS
INDOOR AIR QUALITY IMPROVEMENT GUIDE
INSULATION FACT SHEET- DOE
INSULATION IDENTIFICATION GUIDE
INSULATION R-Values & Properties
METAL LATH, PLASTER & STUCCO
METHANE GAS SOURCES
MOISTURE CONTROL in BUILDINGS
MOLD: A COMPLETE GUIDE TO MOLD
Museum Artifact Preservation
NOISE / SOUND DIAGNOSIS & CURE
ODORS GASES SMELLS, DIAGNOSIS & CURE
PAINT FALURE, DIAGNOSIS, CURE, PREVENTION
PLASTER & BEAVERBOARD & DRYWALL
PLASTER BULGES & PILLOWS
PLASTER LATH, METAL
PLASTER, LOOSE FALL HAZARDS
PLASTER TYPE IDENTIFICATION
PLASTER VENEER Best Practices
ROOF VENTILATION SPECIFICATIONS
SAFETY HAZARDS & INSPECTIONS
SEARS KIT HOUSES
SOUND CONTROL in buildings
SPLITS & CRACKS in STRUCTURAL WOOD BEAMS
STAIN DIAGNOSIS on BUILDING EXTERIORS
STAIN DIAGNOSIS on BUILDING INTERIORS
STUCCO WALL METHODS & INSTALLATION
STRUCTURAL DAMAGE PROBING
SWEATING (CONDENSATION) on PIPES, TANKS
Thermal Expansion Cracking of Brick
THERMAL EXPANSION of MATERIALS
THERMAL IMAGING, THERMOGRAPHY
THERMAL MASS in BUILDINGS
THERMAL TRACKING Indicates Heat Loss
VAPOR BARRIERS & CONDENSATION in BUILDINGS
VENTILATION in BUILDINGS
VINYL CHLORIDE HEALTH INFO
VOCs VOLATILE ORGANIC COMPOUNDS
WALL CONSTRUCTION BARRIER vs CAVITY
WATER ENTRY in buildings
WINDOWS & DOORS
WINTERIZE A BUILDING
WOOD Burning Heaters Fireplaces Stoves
WORLD TRADE CENTER 9-11 DUST PHOTOS
Asbestos test lab procedures, photographs, sample preparation, microscopy methods & example lab report: this article provides photos and procedural suggestions used in the forensic laboratory to identify asbestos materials (or probable-asbestos) in flooring and floor tiles.
Green links show where you are. © Copyright 2014 InspectApedia.com, All Rights Reserved.
Asbestos Test Procedures: Microscopy to Identify Asphalt-Asbestos or Vinyl-Asbestos Floor Tiles in the Laboratory
In this article series we provide photographs and descriptive text of asbestos insulation and other asbestos-containing products to permit identification of definite, probable, or possible asbestos materials in buildings. Readers should also see FLOOR TILE HISTORY & INGREDIENTS for a discussion of the ingredients and production of asbestos-containing flooring.
[Click to enlarge any image]
And see ASBESTOS FLOOR TILE IDENTIFICATION for advice on visual identification of vinyl-asbestos floor tiles or flooring products that probably do or don't contain asbestos.
For a detailed photo guide to individual vinyl-asbestos floor tile patterns, sizes, and years of manufacture, see ASBESTOS FLOOR TILE IDENTIFICATION For a strategy for collecting building dust samples, when, where, how many samples to collect, see DUST SAMPLING PROCEDURE.
While an expert lab test using polarized light microscopy is necessary to reliably identify the presence and specific type of asbestos fiber, or to identify the presence of asbestos in air or dust samples, many asbestos-containing building products not only are obvious and easy to recognize, but since there were not other look-alike products that were not asbestos, a visual identification of this material can be virtually a certainty in many cases.
On occasion, the original flooring packaging or installation literature may be available for a given home: often an extra box of floor tiles was kept for future repairs. The vinyl-asbestos floor tile package label information, combined with a simple comparison of tiles in the package with tiles installed in the building may be sound confirmation of asbestos-containing materials. See Vinyl Asbestos Floor Tile Packaging. Historical information about the dates of flooring installation may also be sufficient to rule in or out the possibility that flooring in a building contains asbestos.
Walter McCrone developed and amply documented the forensic microscopy procedures used to identify asbestos in products, air or dust samples.
Certainly asbestos-certified labs who process large volumes of asbestos samples have developed efficient, high-speed procedures to keep the sample analysis costs down, and surely some of those experts have other tips and ideas for effective processing of floor tile samples besides what we will document here.
However on occasion we need to work with less sample material, or very small asbestos floor tile sample fragments in our laboratory. Here we document and illustrate some suggestions for working with small fragments of vinyl-asbestos floor tiles to obtain material for microscopic examination in the laboratory.
Preparing a Small Floor Tile Fragment for Microscopic Examination for Asbestos
In the lab, following Walter McCrone's procedure for teasing out asbestos particles from solid materials such as this floor tile, we broke a small corner off for further examination by microscope.
Tiles are broken, not cut, in order to expose asbestos fibers for removal, slide preparation, and microscopic examination using transmitted, reflected, and primarily polarized-light central stop diffusion microscopy.
This stereo-microscopic view of the edge of this asbestos-floor tile shows the combination of binder and other silicate materials.
Details about Montgomery Ward vinyl asbestos tile flooring are at MONTGOMERY WARD ASBESTOS FLOOR TILE IDENTIFICATION.
Below (left) is a microphotograph of materials (probably limestone filler) scraped from the broken edge of the Wards vinyl asbestos floor tile shown above. And below (right) is a 1200x magnification photo taken in our laboratory, showing asbestos fibers teased out of the broken edge of a separate sample of floor tile tentatively identified as 2mm x 9"x9" Armstrong Excelon vinyl asbestos flooring ca 1954-1980).
Because many fibers such as fiberglass and asbestos can be almost impossible to detect microscopically, especially in small fragments, unless they are mounted in a proper medium, Cargille certified refractive index liquids (e.g. n=1.550 or n=1.680) are used to mount asbestos-suspect fibers for microscopic examination.
Although McCrone instructs the technician to "tease out" fibers from the edge of a broken floor tile fragment, the "teasing out process" can be tricky if like the author (DF) you have stubby fingers and the sample is about 1cm square.
We use a very small quantity of fast-setting glue to bond our floor tile fragment to a clean microscope slide.
To the right of the floor tile fragment you'll see a drop of clean mounting fluid. We're using Cargille™ N=1.680 in this example. We'll use the droplet to secure fibres we remove from the sample.
Watch out: don't make the droplet too big or you'll waste time looking for your sample fibers under the microscope; don't place the droplet too close to the glued sample or you may overload it with granular debris during fiber removal; don't place the droplet too far away from the sample or you may lose a fiber during transfer from the sample to the drop.
At above left we show our probe under the stereo microscope, as we gently pick away material from the exposed edge of the glued floor tile sample fragment in order to expose a cluster of fibers for further examination.
Our photo at above right shows that we check our mounting fluid droplet for obvious extraneous debris before using it. If the droplet becomes debris-loaded, it's easy to clean the slide and start with a new drop since we've glued down our sample.
At below left a fiber cluster has been removed from the sample and carried to the nearby droplet of mounting fluid. We did not try to remove all of the debris from our fiber cluster as we wanted to keep the fiber bundle intact. At below right, we still haven't lost the sample as we further prepare the slide with a cover slip. If we are using the same slide and glued-sample to prepare a sequence of fiber trials at different Cargille liquid values, we note the refractive index on the slide so as to keep our lab data accurate.
The following six asbestos microscope photos, below show the asbestos fiber cluster from our vinyl asbestos floor tile under magnification and at different lighting conditions.
Our last photo (above right) show mineral fragments from the sample, possibly limestone.
Green link shows where you are in this article series.
Frequently Asked Questions (FAQs)
No FAQs have been posted for this page. Try the search box below or CONTACT US by email if you cannot find the answer you need at InspectApedia.
Use the "Click to Show or Hide FAQs" link just above to see recently-posted questions, comments, replies, try the search box just below, or if you prefer, post a question or comment in the Comments box below and we will respond promptly.
Search the InspectApedia website
HTML Comment Box is loading comments...
Technical Reviewers & References